Manage Out-Of-Tolerance (OOT) Equipment
Manage & Investigate OOT Equipment
When performing calibrations on measuring equipment, it is proper to check the condition of the equipment (aka unit-under-test or UUT) before calibration
(As Found) and after calibration (As Left). If measurement equipment is found out-of-tolerance (OOT) before performing calibration, an assessment of
the effect of the OOT condition on processes must be performed. If it has been 12 months since the last calibration of the UUT, all the product and processes
the UUT measured in the last 12 months are suspect. For instance, were products or processes falsely judged good when they were bad?
In a Calibration record, if the As Found field is OOT, the 'OOT Investigated'
checkbox and the [Print OOT Form] button become visible. Upon finding OOT equipment,
conduct an investigation in accordance with your organization's corrective and preventive
action (CAPA) or nonconforming material (NCM) processes. After the investigation is
complete, check the 'OOT Investigated' checkbox.
If useful to your organization's CAPA or NCM processes, print the OOT Form from the Calibration
record for use as an investigative tool.
Calibration Events with pending OOT investigations are highlighted red in the
Equipment (Tool) Browse when the 'Highlight - Received Out of Tolerance' option
is enabled from the context menu. Clear the red highlighting by checking
the 'OOT Investigated' checkbox for all Calibration Events (see above) that have
OOT conditions for a given Equipment record.
Last updated: 21 Dec 2016
Compatibility: Calibration Control versions 7.x and Above
Download Calibration Control